The Boat Anchor Manual Archive Path: /hp/a/. This is the BAMA archive. These manuals are available for download and free of charge. I am now accepting. View and Download HP A operating and service manual online. DIGITAL VOLTMETER. A Measuring Instruments pdf manual download. Hp A Pdf User Manuals. View online or download Hp A Operating And Service Manual.
|Published (Last):||19 November 2013|
|PDF File Size:||20.27 Mb|
|ePub File Size:||17.27 Mb|
|Price:||Free* [*Free Regsitration Required]|
Outguard Logic Troubleshooting Schematic 8, 9, 10 8-F This allows common-mode mwnual to flow through lead resistance Rb causing some measurement error. Only the system controller can send the Abort message.
Adjustment Procedure A resistor with -m an actual value of ohms would give a measurement result of: The operational verification test should be performed as an incoming inspection of the instrument. ECMR is 3455aa ratio of the peak common-mode voltage to the resultant peak error voltage in the reading with I k l un- balance in low lead.
3455A 5 or 6-Digit Digital Multimeter/Voltmeter
Remove the original AlORl 10 Figure The feedback path for the1 10 Mf2 Reference Measurements. However, the device must be HP-IB compatible. Buyer shall prepay shipping charges to -hp- and -hp- shall pay shipping charges to return the product to Buyer. Rack Mounting Kit Option Jumper Removal A15 Board Remote Permits selected devices to be set to remote operation, allowing parameters mxnual device cKaracteris- tics to be controlled by Bus Messages. The detailed description describes the methods and pertinent circuitry used to accomplish the function of each block of the detailed block diagram.
This item will be shipped through the Global Shipping Program and includes international tracking. Messages are the means by which devices ex- change control and measurement information.
Hewlett Packard HP a 6-digit Digital Voltmeter | eBay
Reinstall the top inner cover with two or three screws and reinstall the lop outer cover bottom covers must be installed. And the voltage reference: If a SA input voltage greater than V is needed, the following procedure manua, always be followed.
Learn how your comment data is processed. The front panel is well lit. The error equation becomes: For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab This amount includes applicable customs duties, taxes, brokerage and other fees.
The cal constant measurement can be aborted a number of times, depending on the function of the instrument.
Low-noise or noiseless, clean ground earth terminal. Each time this button is pressed, the display rate is divided by two. The Input attenuator of the rms converter is an RC circuit which provides a fixed attenuation of to-l on the V ac and V ac ranges. This item will ship to United Statesbut nanual seller has ,anual specified shipping options.
This section contains theory of operation, trouble- shooting procedures, safety considerations, and general service infomation for the Model 34SSA Digital Volt- meter.
At each setting, the A reading should be within the Test Limits given in the table. Figure mankal proper connections for making resistance measurements.
The major portion of communications transmit- ted over the Bus is accomplished by data messages. The full- scale accuracy is therefore: AC Convertor Ranging A Test Oscillator such as the -hp- Model A can be used to verify the ac voltmeter accuracy of the A for frequencies above kHz specified for 1 V and 10 V ranges only.
Agilent / HP 3455A Digital Voltmeter
DC Accuracy Test The input mea- surement is taken at this time. At each frequency setting, the 34S5A display reading should be within the Test Limits given in the table.
The AC Calibrator’s error measure- ment control should then be used to adjust out the kHz errors indicated on the calibration chart. The A reading should be within the test limits listed in Tableverifying its volt tenth scale accuracy with High Resolution on.
When the input voltage exceeds 1. Set the Decade Resistor to The Model A measures dc voltage from 1 dummy cal constant calculation made in the outguard microvolt to KXX volts in five ranges extending from. Typical Operating Charaaeristics A terminal marked with this symbol must be connected to ground in the manner described in the installation opierating mahual, and before operating the equipment. Equipment Required